Time-Compressing Infrasonic Recordings to Discover New Sounds, by Clark Huckaby
Optical Probe: Circuit Description and ApplicationsThe Optical Probe uses a TSL230R
programmable light-to-frequency converter IC made by Texas Advanced
Optoelectronic Solutions (TAOS; now ams). I purchased this chip from Parallax in 2007 but did not find it in their product pages recently (September, 2013); a data sheet is at http://www.c-stamp.com/Documents/Documentation/CS454000d1.pdf. A photo of the Optical Probe assembly is shown in Figure 5 of the Overview. The "chassis" is a 3.75-inch length of 3/4-inch wide
right-angle aluminum stock. At one end, a male XLR jack (J1) is pin-compatible with
the Interface Unit, so a standard mic cable can be used for the connection (avoiding excessive length). The other end holds a 0.6
X 2.2-inch perfboard with the TSL230R and other components wired point-to-point.
schematic is shown below, in Figure 1 (figure and component numbers apply to this page only). The recommended nominal supply voltage (VDD) for the TSL230R (U1) is +5 VDC,
so I used this as the Interface Unit's
supply voltage. Logic inputs at pins 1 and 2 of U1 set the
relative light sensitivity to 1X, 10X, or 100X (in effect an "aperture
control" enabling 1, 10, or 100 percent of the device's photodiode
array area), or activates a sleep mode. Pins 7 and 8 program on-chip
frequency division by 1, 2, 10, or
100. With pull-down resistors R1 through R4, switch array SW1
these operating modes (see tables on the right-hand side of Figure 1); the output duty cycle is 50 percent except at unity
Figure 1. Schematic diagram of Optical Probe (left). Tables showing operating modes as determined by SW1 setting (right).
As characterized in Figure 1 of the TSL230 datasheet, output frequency is a positive linear function of light intensity (irradiance in units of power per area illuminated [µW/cm2])
across six decades for any given frequency divisor setting. Overall,
the device's linear response spans at least 10 decades considering
all available sensitivity and frequency division settings. However, as mentioned in the Data Converter description,
my recording channel's phase-locked loop (PLL) limits the maximum signal
amplitude, or relative light intensity range in this case, to less than
one decade (about 200 to 1800 Hz at the
PLL input, in frequency terms).
(Note that any frequency analysis system [or demodulator] has a limited
dynamic range. For example, frequency measurement requires at least as
much time as the
reciprocal of the lowest frequency [i.e., the period]. This effectively ties
channel bandwidth to amplitude at the low end.)
phenomena have periodic light intensity changes that exceed the
amplitude range of my Infrasonicon channel. As noted in the Overview Part 3,
in such cases the Optical Probe can be "photobiased" (illuminated) with
a LED operating at constant current. This is analogous to lowering the
contrast of an image projected on a screen by turning on room
lights. Normally, when modulation exceeds the 200-1800 Hz range at the Data Converter's
frequency input, photobias intensity (LED current) is increased and the
resulting increase in FM carrier frequency compensated by increasing
the frequency divisor (either at SW1 or in the Interface Unit).
When modulation is within the safe range, fine adjustment of the
photobias LED may be used to set the DC offset of the demodulated
Examples of recordings that required photobias are presented at http://clarkhuckaby.com/NewVibe/CloseLookVibe.html, where I investigated neon lamp activity in Fender optical tremolo circuits. Figures 2B and 2E on that page are photos of Infrasonicon and Optical Probe implementation in those experiments, and Figures 5 and 6
show recorded neon light pulse shapes (waveforms). Photobiasing was
obviously necessary because the neon lamp has a dark phase
in each cycle. The tremolo experiments also exemplify how, given its
linearity, the Infrasonicon channel can be an analytical tool
(oscilloscope) as well as a means to discover new sounds.
Overview Part 3 has more about the Optical Probe, along with links to example sounds.
Infrasonicon Home Page
Infrasonicon Overview Sound Example Page
Huckaby Fender Optical Tremolo Study
Go to clarkhuckaby.com
Copyright 2013 Clark
Huckaby. All rights reserved. Commercial use of this content (text, images,
sounds, or source code) without first obtaining the expressed
permission of Clark Huckaby is prohibited by
law. Distribution of copies beyond personal use and/or re-posting any or
all of this page is prohibited by law.